Often times, a high impedance reference electrode can lead to your potentiostat oscillation. More information on why can be found in our potentiostat hardware manuals. In today’s tech tip though, we’re going to highlight three methods to dealing with a high impedance reference electrode.
Frequently Asked Questions
Setup & Use of Gamry Products
Answers to questions that go beyond setup and basic device information.
Authorization Codes activate your Gamry Software to work with your potentiostat. Each of the software packages has a package name (e.g. “DC105” for DC Corrosion Techniques) and a 10-digit Authorization Code that is linked to the serial number of your instrument.
After updating Framework you may be prompted to update the firmware on your instruments. To begin, make sure the instruments that need a firmware update are powered on and connected to the computer via the USB cable. This guide is valid for all Reference and Interface family instruments
We have seen several examples where a customer has recorded an open circuit voltage of 3, 4, or even 5 volts when running EIS on a coated sample. The Eoc values that were recorded in these experiments are probably not real Eoc readings. For very, very good coatings, such as the ones that these researchers were studying, we recommended that the potentiostatic EIS measurement be done at a potential vs. Eref, not vs. Eoc. The Eoc values are often not related to the coating or to the substrate: They are a function of the length of time the potentiostat has been connected to the cell.
The “Copy to clipboard” button is located on the toolbar of the Chart tab when your data file is displayed in Echem Analyst. This button gives you the option to copy a Bitmap, Metafile, or raw data directly to the Windows Clipboard. It can then be pasted into any other program. The Bitmap option will copy the data as displayed in Echem Analyst, meaning that it will be a replica of the current view of the data complete with any overlays, fits, or modifications of the visual style of the plot. Using the “As Text (data only)” button will copy the X and Y values of the currently displayed data, fits, and overlays as a table that can be directly pasted into a cell-based spreadsheet application. By overlaying multiple data files you can use the Copy As Text feature to batch-export a number of data files in one fell swoop.
There are several possible reasons that this message will appear during an EIS scan. The first thing to check is that all of your cell connections are appropriate. If the error continues to appear one should run a full calibration of the system. During the calibration the performance of the EIS components are checked. Depending on the results of the calibration, we can determine if the problem may be cable related or if a repair of the instrument is required. Please email your Calibration Results file (generated during calibration and stored as a .txt file in the My Gamry Data folder) to Gamry by using the Technical Support Form.
Microsoft is Ending Support for Windows 7
Microsoft will discontinue support for Windows 7 on January 14, 2020 which means Gamry will also be discontinuing support for Windows 7. If you are upgrading to Windows 10, like many companies and institutions, you need to be running Version 7 of our software. Please note that only USB and Ethernet-based instruments can run in Version 7. Eligible users can download the latest version of our software through our online Client Portal.
If you haven’t already registered your instrument, you can do so through the Client Portal.
Please email Technical Support if you have any questions regarding this transition. Please be sure to include your instrument model and serial number when contacting us.