This instance function is available for all signals.
Type of acquisition mode:
0 = Fast Mode
1 = DSP Mode
Use the default AcqMode. This mode is set based on the output point sample period:
Below 100.0 µs the system uses Fast Mode.
At or above 100.0 µs, the system uses DSP Mode.
|AcquBasePeriod||REAL||The time between each sample point. This number should not be less than the default of 16.666 µs.|
Use the default AcqBasePeriod.
For ACQ_MODE_FAST, this is not used.
For ACQ_MODE_DSP, the default is 16.666 µs.
|SampleRate||REAL||The output point rate, in seconds. Output points are the points displayed to the user and saved to the curve. They are made up of one or more sample points.|
|NIL||Use the default sample rate specified in the SIGNAL.New call.|
|DutyCycle||REAL||A number between from 0.0 to 1.0 specifying the duty cycle that is used in sampling. 0.0 always gives one sample, while 1.0 samples as quickly as possible.|
0 = One sample only (duty cycle of 0.0)
1 = Use the default duty cycle of 0.20
2 = Use a duty cycle of 1.0
|NIL||Use the default duty cycle of 0.20|
NOTE: For a PCI4 family potentiostat the parameters are ignored. Only the Fast Mode is available.